Crenova MS8233D Multimeter is an Auto-Ranging Digital Multimeter *The Digital Multimeter, as general measurement tool, is widely used in the school, laboratory, factory and other social field. *The equipment is used for measurement in building facilities. For examples, distribution boards, circuit breaker and industrial equipment located in fixed facilities, as a fixed motor. Crenova MS8233D Multimeter is very easy to use *The convenient auto-ranging operation allows you to select the right measurement range when working on a variety of components, from small electronics to main electrical sources. *Very easy to read labels and screen makes it easy to select desired tests and the results. Technical Specifications: 1.DC Voltage: *200mV/2V/20V/200V ±(0.5%+2) *600V ±(0.8%+2) 2.AC Voltage: *2V/20V/200V ±(0.9%+3) *600V ±(1.2%+3) 3.DC Current: *200?A/2000?A/20mA/200mA/10A ±(1.5%+3) 4.AC Current: *200?A/2000?A/20mA/200mA/10A ±(1.5%+4) 5.Resistance: *200?/2k?/20k?/200k? ±(0.8%+2) *2M?/20M? ±(1.0%+2) 6.Frequency: *20kHz ±(1.5%+5) 7.Power Supply: 9V Battery (6F22 or 1604A Type) 8.Safety Class: IEC 61010-1, CAT II 600V. Safe and Reliable *Over Load Protection: Use the PTC protection circuit for Resistance, and frequency measurement. *Auto Power OFF: If the meter is idle for 15 minutes (idle time), the meter will automatically turn off. Warning Tips: 1.Please install a new battery before using Step One: Remove the back cover screw Step Two: Open the back cover and install the 9V battery 2.Please read relevant information carefully and observe all the warnings and notes strictly. The meter can be cleaned with soft clean cloth to remove any oil, grease or grim. Do not use liquid solvent or detergent.
*Multimeter Over load protection:Use the PTC protection circuit for Resistance, and frequency measurement.
*Digital Multimeter 3 1/2 digit LCD display with back light makes the reading clear.
*Electrionic meter can measure AC/DC voltage, AC/DC current, resistance, frequency, continuity, diode and etc.
*Measuring Instrument with function of data hold, Maximum Value hold, backlight, function switching.
*Portable multimeter,Auto Power OFF: If the meter is idle for 15 minutes (idle time), the meter automatically turns off. After auto power-off, pushing any of the push button or changing the rotary switch can turn on the meter again. (Note: it needs install a new battery before using; there maybe a little corlor difference)